Records of periodic spot-checks for remote afterloader units.
shall retain a record of each spot-check for remote afterloader units required
of the Administrative Code for three years.
The record must include, as applicable:
The date of the spot-check;
The manufacturer's name, model number,
and serial number for the remote afterloader unit and source;
An assessment of timer accuracy;
Notations indicating the
operability of each entrance door electrical interlock, radiation monitors,
source exposure indicator lights, viewing and intercom systems, and clock and
decayed source activity in the unit's computer; and
The name of the individual who performed
the periodic spot-check and the signature of the authorized medical physicist
who reviewed the record of the spot-check.
A licensee shall retain a copy of the
procedures required by paragraph (B) of rule
of the Administrative Code until the licensee no longer possesses the remote
Five Year Review (FYR) Dates:
Prior Effective Dates:
Prior History: (R.C.
119.032 review dates:
Prior Effective Dates: 8/15/2005