(A) Before the first medical use of a brachytherapy source, a licensee shall have:
(2) Determined source positioning accuracy within applicators; and
(3) Used published protocols currently accepted by nationally recognized bodies to meet the requirements of paragraphs (A)(1) and (A)(2) of this rule.
(B) Instead of a licensee making its own measurements as required in paragraph (A) of this rule, the licensee may use measurements provided by the source manufacturer or by a calibration laboratory accredited by the "American Association of Physicists in Medicine" (AAPM) that are made in accordance with paragraph (A) of this rule.
(C) A licensee shall mathematically correct the outputs or activities determined in paragraph (A) of this rule for radioactive decay at intervals consistent with one percent radioactive decay.
Prior History: (Effective:
R.C. 119.032 review dates: 06/28/2010 and 07/12/2015
Promulgated Under: 119.03
Statutory Authority: 3748.02, 3748.04
Rule Amplifies: 3748.04
Prior Effective Dates: 8/15/2005 )