(A) A licensee shall have each teletherapy unit and gamma stereotactic radiosurgery unit fully inspected and serviced during source replacement or at intervals not to exceed five years, whichever comes first, to assure proper functioning of the source exposure mechanism.
(B) This inspection and servicing may only be performed by persons specifically licensed to do so by the director, the United States nuclear regulatory commission, or an agreement state.
Five Year Review (FYR) Dates: 05/26/2015 and 06/01/2020
Promulgated Under: 119.03
Statutory Authority: 3748.02, 3748.04
Rule Amplifies: 3748.04
Prior Effective Dates: 8/15/2005, 10/4/10