The licensee shall perform acceptance testing on the treatment planning system of therapy-related computer systems in accordance with published protocols accepted by nationally recognized bodies. At a minimum, the acceptance testing must include, as applicable, verification of:
(A) The source-specific input parameters required by the dose calculation algorithm;
(B) The accuracy of dose, dwell time, and treatment time calculations at representative points;
(C) The accuracy of isodose plots and graphic displays;
(D) The accuracy of the software used to determine sealed source positions from radiographic images; and
(E) The accuracy of electronic transfer of the treatment delivery parameters to the treatment delivery unit from the treatment planning system.
Prior History: (R.C.
119.032 review dates:
Promulgated Under: 119.03
Statutory Authority: 3748.02, 3748.04
Rule Amplifies: 3748.04
Prior Effective Dates: 8/15/2005 )