Rule 3701:1-58-50 | Therapy-related computer systems for manual brachytherapy.
The licensee shall perform acceptance testing on the treatment planning system of therapy-related computer systems in accordance with published protocols accepted by nationally recognized bodies. At a minimum, the acceptance testing must include, as applicable, verification of:
(A) The source-specific input parameters required by the dose calculation algorithm;
(B) The accuracy of dose, dwell time, and treatment time calculations at representative points;
(C) The accuracy of isodose plots and graphic displays; and
(D) The accuracy of the software used to determine sealed source positions from radiographic images.
Last updated November 18, 2022 at 8:36 AM